Thinklabs One User Manual Thinklabs !!LINK!!
Please call to obtain a Return Authorization (RA) number within 30 days from the date of the invoice prior to returning any merchandise. Items must be in new and resalable condition in the original packaging with all necessary components including the user manual. Return may be subject to a restocking fee. Non-defective product showing signs of use is not returnable.
Thinklabs One User Manual Thinklabs
The stethoscope was designed for high power output, with dedicated headphone amplifiers and has fully adjustable volume levels (1-10). The device can produce very high sound pressure levels and can exceed 100 times the decibel level of a conventional stethoscope. Five different filters allow the user to switch between listening for heart and lung sounds. In addition to filtering operations, adjusting the amount of pressure applied to the diaphragm alters the frequency response. Filter one ranges from 30 to 500Hz and filter two ranges from 60 to 500Hz, these low-pitched filters provide optimal auscultation for heart sounds. Filter three ranges from 80 to 500Hz and filter four ranges from 100 to 1000Hz, these filters are at a higher pitch, providing optimal auscultation for lung sounds. The fifth filter is a broadband filter ranging from 20Hz to 2000Hz +/- 3dB. The broad range of this filter lacks the specificity to filter out ambient noise but is ideal for recordings. The unit allows providers to select two filters as their "favorites" and set them for easy transition between the two filters. The LED lights are to serve as a visual indicator that the stethoscope is functioning. These LEDs also serve to indicate the volume level, current filter selection, and the device's battery life. The stethoscope contains a rechargeable lithium-ion battery that can be recharged through a USB port. A full battery charge provides four continuous hours of auscultation time, which provides 100+ patient assessments. To save battery life, it e has a customizable timer shut off that can be set between 1 and 10 minutes. If the user prefers, the stethoscope can be placed in a continuous "on" mode where the provider can manually turn the stethoscope off when not in use. It is designed to resemble the tactile shape and features of a traditional stethoscope, making it intuitive for users. The device was designed to be able to create audio recordings while auscultating. The stethoscope is compatible with both Mac and Android software. The manufacturer has designed an application (app) that allows audio recordings to be made with visual indicators.4